Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Frequency Counter - Event Counting (Total)
All events meeting the Trigger Slope, Level, and Hysteresis criteria on the selected Trigger Source channel are counted. This includes trigger events that cause a trace update, additional events within a trace, and events that are not shown because they arrive between trace updates. Normal trigger mode is usually best, though Auto and Auto Level may sometimes be useful.
Alternatively, Gen Sync may be used instead of one of the above Trigger modes, for example to count the total number of Generator tone bursts with Burst Sync (or Train Sync), or the number of sweeps with Sweep Sync.
Trigger Delay may be used to position the event as desired within the trace area; it has no effect on the Total.
For events that are bursts of tone or noise, you usually want to count only the start of each event, not every individual cycle or noise excursion within the event. Trigger Holdoff allows you to specify a "dead time" during which possible trigger events are ignored; if you set it to a value somewhat longer than the burst, then the trigger detector will be locked out for the duration of the burst after the initial starting event.
Total mode can count events that are distinguished by a simple level threshold, such as a burst of noise. If you need to distinguish between bursts based upon their spectra, consider SpecTot mode.
FcountMode=Tot or FcountMode=3 sets the Frequency Counter to Total mode. See Macro Notes under Frequency Counter - Hertz Measurement for the complete list of mode names and numbers.
Freq is an internal variable that holds the current Frequency Counter readout value, dependent on the chosen mode (Hz, RPM, msec, Total, or SpecTot).
Total holds the count of total cycles counted since the last Reset, valid during Hz, RPM, or msec modes. This allows macros to get the Total value even when these other modes are active. Note that this does not work in SpecTot mode.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 30 Years of Innovative Instrumentation
© Copyright 2007 - 2017 by Interstellar Research
All rights reserved