Daqarta
Data AcQuisition And RealTime Analysis
Scope  Spectrum  Spectrogram  Signal Generator
Software for Windows Science with your Sound Card! 

The following is from the Daqarta Help system:

Features:OscilloscopeSpectrum Analyzer 8Channel

Applications:Frequency responseDistortion measurementSpeech and musicMicrophone calibrationLoudspeaker testAuditory phenomenaMusical instrument tuningAnimal soundEvoked potentialsRotating machineryAutomotiveProduct testContact us about


Voltmeter RMS Controls: Options >> Voltmeter >> RMS
The Voltmeter RMS option computes the true Root Mean Square value of the waveform, over an integer number of waveform cycles. RMS is computed by taking the square of each data point, summing all the squares together, dividing by the number of data points, and then taking the square root of the result. The RMS computation assumes that enough data points are included to get a good representation of the average value of the signal. For lowfrequency repetitive waveforms, highest accuracy is obtained when the computation includes an integer number of cycles. In order to obtain this, the trace must be Triggered and the 1024 trace samples must include at least one full cycle of the waveform. The lowest frequency you can measure this way is thus the sample rate divided by 1024: 96000 93.750 Hz 48000 46.875 44100 43.066 22050 21.533 11025 10.767 For stilllower frequencies, all 1024 samples are used and the reported value may fluctuate over time, as the readout shows the average value for any given 1024 samples. All 1024 samples are also used if Trigger is off. Since there will not in general be an integer number of cycles, the accuracy of the reading may be reduced for low frequencies. Note that for a symmetrical wave, an integer number of halfcycles is as good as an integer number of cycles for ultimate accuracy. The worst case, assuming more than one cycle is present, is when there are 1.25 cycles included; the instantaneous error may be several percent. However, the displayed value is further smoothed using the same time constant (TC) as for trace cursors, so the reading will usually quickly settle down to an error of less than one percent using the default TC of 5. You can increase the effective averaging time by increasing the TC. As the signal frequency increases, the error due to noninteger cycles diminishes because any fractional cycle is a smaller part of the total. Random noise measurements should always be made with Trigger off. The RMS option can be used with file data, but since there is no Trigger available for singletrace files, only long (DDisk) files can use Trigger to get the most accurate reading... even though only waveforms that fit into 1024 samples will be able to take advantage of this. Note that the RMS Voltmeter option is essentially the same process as the Sigma cursor readout option applied to the waveform, but RMS Voltmeter uses Trigger information instead of manual cursor positions to automatically set the optimal summation region. For signal or noise bursts, note that the true RMS value (effective heating energy) must consider the overall cycle time of the burst, including the dead time between bursts. For burst cycles that are shorter than 1024 samples, the RMS reading will be correct because every calculation will include at least one full cycle. That will be true whether Trigger is active or not. However, for longer bursts or longer cycle times, you will need to turn Trigger off and rely upon the averaging effect of the Voltmeter Time Constant to get the true RMS value. This depends upon bursts falling randomly within the 1024sample data set of each update; some updates may see only the On portion of the burst, and some may see only the Off portion, but the average of many updates should yield the true RMS value. Macro Notes: VoltMode=RMS or VoltMode=0 sets the Voltmeter to RMS mode. You can use either mnemonic names or numeric values (or variables or expressions) for setting, reading, or IF testing: 0 = RMS 1 = Peak 2 = Pos 3 = Neg 4 = dB Note that if you use a name instead of a value in an IF test, it must be in quotes: IF.VoltMode="RMS". Posn?S is a readonly macro that reports the samples used for RMS measurements. Note that this value may include a fractional part due to interpolation to find the effective (fractional) sample number where the signal would have exactly hit the Trigger Level between integer sample numbers. The value is rounded off by the Voltmeter to get the actual number of samples to be included in the computation. Note that this value may be read even if the Voltmeter is not active, or is active but using some other mode. A macro can use this to find the waveformsynchronous RMS value for any channel, even a channel other than the triggered channel, as long as it is correlated with the triggered channel. This is useful for RMS measurements of inputchannel responses to outputchannel driving signals, where the response may include noise or interference that would cause triggering difficulties. Instead, you trigger on the "clean" output channel and measure on the "dirty" response channel. For example, assuming the trigger parameters have been set for the desired output channel, and you want to measure the response on Left In (channel 0), you could use: Buf5="<=W0" ;Copy Left In to Buf5 (arbitrary) Ch=5 ;Set Buf5 channel select for BwSig() V=BwSig(0,Posn?S1) ;Sigma function returns V = RMS Notice that the BwSig Sigma function includes Posn?S samples (0 to Posn?S1), but doesn't actually care if the 0th sample was the first sample in a waveform cycle; as long as it includes an integer number of cycles, and the waveform is repetitive, it can start on any point in the waveform cycle. Posn?S is used by the _Multi_Mtr_Task of the Multiple Meters miniapp for RMS readings in Main Trig Sync mode. See also Voltmeter, Voltmeter Channel 

GO:
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.INTERSTELLAR RESEARCH: Over 30 Years of Innovative Instrumentation © Copyright 2007  2017 by Interstellar Research All rights reserved 