Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Spectrum Cursor Track Dialog
This button opens the Spectrum Track dialog, which allows the solid cursor to be automatically positioned over the largest spectral peak. The button that toggles the Track option is at the top of the dialog. There are also controls that allow you to limit the tracking to peaks above a specified threshold, or between specified maximum and minimum frequencies.
If the signal has harmonics present, you can elect to track the fundamental instead of harmonic peaks, even if the harmonics are larger or the fundamental is missing completely. You can also elect to have the dotted cursor track a specified harmonic.
Note that peak height, and hence Track operation, is affected by any active Spectrum Curve weighting. That is only active if Spectrum Y-log (dB) mode is also active, even if you are currently in Pitch Track mode where dB is always used.
When the Spectrum control dialog is open, SpectTrackDlg=1 opens the Spectrum Track dialog, SpectTrackDlg=0 closes it, and SpectTrackDlg=x toggles between open and closed.
Alternatively, when the Spectrogram / Pitch Track control dialog is open, SgTrackDlg=1 opens the Spectrum Track dialog, SgTrackDlg=0 closes it, and SgTrackDlg=x toggles between open and closed.
Note that you do not need to open the Spectrum Track dialog to change its controls directly via macro commands.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 30 Years of Innovative Instrumentation
© Copyright 2007 - 2017 by Interstellar Research
All rights reserved