Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Juxt Trace Memory Array
Traces may be recalled individually or viewed as an array. Array traces may be labeled and/or sorted according to the contents of designated Fields, and displayed in ascending or descending order with adjustable spacing. You may also choose to Hide any unwanted traces, such as to simplify comparisons.
Juxt Array Usage - Quick Guide
In a typical application, you acquire a trace, Pause, and hit the Juxt New button to save it. (Or hit ALT+J to save directly, without a separate Pause step.) Then you change some test parameter (loudness or frequency, say) and acquire and save another trace. Toggling the Array button on will show all the saved traces on the screen at once.
Using the mouse, you can adjust the spacing between the traces by holding the ALT key and dragging them apart or together, and adjust the overall vertical position by holding the SHIFT key and dragging the array up or down the screen.
Measurements can be made on an individual trace by selecting it with a left click. That trace will be highlighted, and all the cursor readouts, Notes, Labels, Fields, and dialog settings like total averager frames will pertain to that trace.
To alleviate clutter and simplify comparisons, CTRL plus a left click on any trace will cause it to be hidden. The Show button will show all the traces, with the formerly-hidden ones shown dotted. CTRL-clicking on any trace will then toggle between hidden and normal, so you can choose different traces to be compared and then toggle Show off again.
Juxt Array Dialog
CTRL+J opens the Juxt Array dialog with the following controls:
Note that the Just Array dialog may not be opened during DDisk recording, though you can still save traces to the array... you just can't review them or save the array to a file until the recording is complete.
JuxtDlg=1 opens the Juxt dialog, JuxtDlg=0 closes it, and JuxtDlg=x toggles between open and closed.
Note that you do not need to open this dialog to change most of its controls directly via macro command, as long as they are enabled.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2021 by Interstellar Research
All rights reserved