Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Revision History - v4.00
Daqarta v4.00.2 - 24 June 2008:
Bug Fix: Daqarta could crash while loading certain Generator setups.
Daqarta v4.00 - 8 June 2008:
Sound Level Meter option added, with six standard and unlimited custom weighting curves, plus five standard or custom adjustable attack and decay time responses. Continuous Peak and PeakTC buttons allow viewing either the current SPL, or the peak SPL since the last Peak Reset, with and without response time constants. Adjustable decimal places for SPL display.
For Pro or trial users, complex tasks can now be fully automated with a scripted Macro system that includes loops, custom status and prompt messages, timers (including the ability to awaken the system from standby), event waits, local variables that persist between sessions, and bounded random value entry to controls. Macros can be invoked via user-defined hot-keys or selected from a list. Macro collections can be saved to and loaded from .MAC files, including an auto-loaded file that can run a custom startup macro automatically. (Non-Pro users, including free expired-trial users, can use all these features except saving macro files. They can use files created before the trials expired, and they can use macros created during the current session.)
For Pro or trial users, the Juxt Array system allows multiple traces to be stored in memory and shown individually or as multi-trace displays, sorted and labeled according to a chosen Field, date, or time. Trace arrays can be saved and loaded as .JXT files.
Voltmeter now shows SPL in dB mode, where applicable. Continuous Peak Hold allows viewing peak or instantaneous values for any mode at any time. Separately-adjustable decimal places for linear and dB modes. Time constant is now adjustable independently from cursor readout TC. Voltmeter now works on all files, formerly only Paused long (DDisk) files.
A Generator Clip Indicator below the main toolbar now shows when instantaneous digital waveform values attempt to exceed full-scale, typically from combining multiple Streams with Level values totalling more than 100%.
Most readouts (Averager Frames counter, cursor readouts, file position, modulation status, Wave type, dB, etc) may now be copied to the clipboard via normal Windows highlight and CTRL+C, and pasted into Fields and elsewhere via CTRL+V. For the Frequency Counter, Voltmeter, and Sound Level Meter, left-clicking anywhere on the displayed value captures it and pops up a message asking if you want to send it to the Clipboard.
Cursor readouts for X position are now controls that allow direct entry, as well as display of positions set via mouse or cursor key.
Pause button now has red highlight "LED" when active.
Default font sizes are set in initial session so Daqarta fills the screen as much as possible. Manual settings take effect for future sessions.
Bug Fix: For Gen Sync Trigger mode, if next trace update would be delayed more than about 5 seconds while waiting for tight Maximum Sync Error, the sync happens on the next cycle within +/-0.5 samples of the target.
Bug Fix: Waveform Average files did not display properly for other than simple mono and stereo channels. Now works with any combination of up to 4 channels (0, 1, or 2 In and 0, 1, or 2 Out).
Bug Fix: Spectrum of Wave Average was not shown properly.
Bug Fix: Home and End keys now work for all edit controls. Also Notes now accepts Tab key and inserts 4 spaces.
Bug Fix: Negative SPL values on the Sgram dB axis were shown the same as positive, now shown in red. For normal relative dB, positive values (rare) are shown with a leading '+' sign.
Bug Fix: HTML Help window could not be repositioned. Possible crash if Daqarta closed while HTML Help was still open. PgUp/PgDn now works in Contents as well as topic pages.
Bug Fix: Auto-Increment Filenames options was improperly incrementing names like "Test327" to "Test328", etc, instead of "Test327-000", "Test327-001", etc.
Bug Fix: Cursor TC formerly allowed values of 1-31, with 1 claimed to be no smoothing at all... but was actually a very small amount. Now allows 0-31 and 0 really is no smoothing.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2021 by Interstellar Research
All rights reserved